Tel, Viber, WhatsApp, Telegram: +38 067 895 98 24
Medical Ventilator SERVO - Mixer Error - Repair.
Medical Ventilators this is a Ventilators for intensive care units, equipped with contemporary and sophisticated ventilation modes, combined with a user-friendly interface (Ventilator ICU).
In the ICU ventilator SERVO, most of the problems are related to the operation of the mixing unit.
The most common cause of errors of the Medical Ventilators SERVO is a dirty filter of the mixer in chamber pressure sensors and an oxygen sensor.
The filter stands on the mixing module air duct on the back wall behind the gas module chambers of the Medical Ventilators.
Medical Ventilator SERVO - mixing modules. Photo.
Medical Ventilator SERVO - filter location. Photo.
Medical Ventilator SERVO - filter element. Photo.
If Medical Ventilator without a filter element gives an error, check the electrolytes on the power supply board. The photo shows electrolytes circled in white.
Medical Ventilator SERVO - power block.
Another article on the Medical Ventilator Hamilton G5 repair:
Hamilton G5 Ventilator - Error 5507 Repair.