Tel, Viber, WhatsApp, Telegram: +38 067 895 98 24
Point Nix 3D - error P5.
Point Nix 3D Combi gives an error P5 if there is no 100kV per x-ray tube.
The tomograph works until the "Scan" button is pressed.
After clicking the Scan button, the Point Nix 3D 500 tomograph is immediately blocked by protection. P5 lights up on the display, all LEDs are flashing.
To restart, you need to turn off and turn on the machine.
The fault may be located in the oil block of the X-ray tube, as well as in the power supply blocks of the Point Nix tomograph.
Articles about repairing the dental scanner a Point Nix 3D Combi:
Point Nix 3D Combi - PS1, PS2, PS3, PS4 errors.
Point Nix 3d Combi 500 no voltage on the CN14 connector.
Point Nix 3D Combi - soft and hard installation.
Pointnix 3d combi 500 power block repair.
Cone-ray tomograph - obtaining 3D images.
Point Nix 3D Combi malfunction definition.